Front InletBack Inlet 29 To configure the Gas type 29 To configure the PTV or COC coolant 29 To configure an MSD transfer line heater 45 To configure a nickel catalyst heater 46 How FID units are displayed in Agilent data systems and on the GC 300 To light the FID flame 300 If using an an Agilent 6890 GC or if performing an installation on an Agilent 7890 GC in a position other than an inlet or detector position, follow this step for connection of the heater cable to SplitSplitless Inlet Illustrated Parts Breakdown.
Headquarters Other sites 5301 Stevens Creek Blvd. Santa Clara, CA. United States The following are the manual steps for performing the standard inlet tests for 6890 and 7890A GCs. Typically, one runs the Inlet Leak Check to confirm. Skip navigation. Home; Manual Steps for Inlet Leak Check, Pressure Decay Test, and Split Vent Restriction Test for Agilent Operating Manual Volume 1. General Information Agilent 6890 Series Gas Chromatograph Manual and parts of G Maintenance and Troubleshooting Manual Printed in USA Warranty The information contained in this document is subject Inlet types The HP 6890 GC has five types of inlets available.
All are offered with electronic pneumatics control (EPC) and two are offered without. Table 1. Inlet Types Advanced Operation Manual 5 COC inlet Agilent 6890 inlet manual transfer control 91 Setting COC inlet flowspressures 92 Setting COC inlet parameters 93 Agilent 7890 Series Gas Chromatograph Advanced Operation Manual Agilent Technologies 1 Programming Run Time Programming 12 Operating Manual Volume 3.
Detectors Agilent 6890 Series Gas Chromatograph. Agilent Technologies, Inc. 2850 Centerville Road Operating Manual Volume 3. Printed in USA The information contained in inlet and detector column fittings are either connected to a column or capped Agilent 6890 Gas Chromatograph Service Manual 420 Oven and Temperature Control Heated zones If a manual inlet carrier is installed, route the cryo tubing out of the slot Agilent 6890 Gas Chromatograph Service Manual Oven and Temperature Control 420.
Agilent Technologies Agilent 6890 Gas Chromatograph Maintaining Your GC. Notices To Attach a Sample Transfer Line to the VI 138 To Remove the VI Interface 139 This manual details the routine tasks needed to maintain the 6890 Gas Chromatograph (GC). The procedures assume a basic 240 Programmed Temperature Vaporization Inlet (PTV) Theory of operation 2 of 46 Inlets Jun 2001 Agilent 6890 Gas Chromatograph Service Manual Sampling heads The septum head uses either a regular septum or a Merlin microseal to seal Agilent 6890N Network Gas Chromatograph Data Sheet Chromatographic Performance with both Agilent 6890N and Agilent ChemStations.
Automated carrier has constant flow, when capillary column dimensions are entered into the 6890. SplitSplitless, PTV inlets have flow sensors for the control of split ratio. Inlet modules